OFFICIAL NATIONAL DISTRIBUTOR
Trident P2 Level 3 Surgical Disposable Face Mask
Each box contains 20 individually packaged masks
Regular Standard Size.
- P2 certified
- AS 4381: 2015 Level 3, 160 mmHg (ID No. BMP 737354)
AS/NZS 1716: 2012 (ID No. BMP 732382)
- Advanced multilayer technology providing protection against fluids, dusts, mists, fumes and smoke
- Three panel design provides structural integrity, form fit seal and optimal comfort. All three mask panels and seams have been independently tested and certified to AS 4381:2015 Level 3 – ISO 22609
- Bacterial Filtration Efficiency (BFE) >99%
- Fluid protection level 3, 160 mmHg
- Proprietary adjustable foam nose pad to ensure consistent optimum facial seal and reduce fogging of eyewear
- Versatile unisex fit suitable for a range of facial shapes and sizes. Effective for many “hard to fit” individuals
- Specialised TRIDENT® face masks seal attains unrivalled PortaCount® fit testing results
- Low breathing resistance, maximizing wearer comfort, durability and life of product
- Designed to be used in conjunction with other mandatory personal protective equipment
- Each mask is individually batch numbered and packaged for the ultimate in traceability, portability and hygiene
Latex free to avoid the risk of developing allergies with repeat exposure
TRIDENT® P2 Surgical White Paper Study
In a recently published “White Paper” comparing high grade mask quantitative fit tests the TRIDENT® P2 achieved outstanding results. Professor Daryl White and his colleagues at Melbourne University recruited 500 participants for their comprehensive study. Their finding were consistent with previous research showing very high QNFT pass rates with three-panel flat-fold respirators. Their tests overwhelmingly showed the TRIDENT ® P2 outperformed their 3M competitor “both the overall and first-attempt QNFT pass rates of the Trident P2 were significantly higher than the 3M (99.2% vs 92.6%, p < 0.001; and 92.6% vs 76.4%, p < 0.001 respectively). The overall fit factor and the individual fit factor were also significantly higher for the TRIDENT FFR than the 3M.
You can access and download a full copy of the White Paper Study as published in “Science Direct” using the link below:
HOW TO APPLY – DONNING
- Position the face mask in the palm of your hand with the nose strip at your fingertips, ensuring both head straps are untwisted over the back of your hand.
- Open both the top and bottom panels of the mask and make a bend in the foam nose seal. Place the lower panel under the chin and the upper panel with the foam nose seal over the nose bridge and underneath the eyes.
- Holding the face mask with one hand, position the top untwisted head strap to the crown of the head.
- Draw the lower untwisted head strap up and over the face to below the ears and back of the neck. DO NOT CRISS CROSS HEADSTRAPS.
- With both hands, mould the foam nose seal at the top of the mask to the nose bridge and cheeks ensuring a secure fit and seal.
- Perform a fit check (see fit check list below). Cover the front of the face mask with both hands. Inhale and exhale sharply.
HOW TO APPLY – DOFFING
- The face mask may be contaminated. Touch only the head straps when removing.
- Pull the bottom strap over the top of the head.
- Pull the top strap over the top of the head.
- Dispose of face mask into the appropriate waste receptacle in accordance with site infection control and prevention policies.
- Wash and sanitise your hands thoroughly.
FIT CHECK LIST
- If air leaks at the nose or cheek areas adjust the foam nose seal until a seal is achieved.
- If air leaks around the edge of the face mask adjust the head straps and reposition the mask until a seal is achieved.
- A successful fit check is when there is no air leakage.
- Always fit check before entering the work area.
- If a successful fit check cannot be achieved, remove and refit the face mask.
- If you still cannot obtain a successful fit check, do not enter the work area.
NOTE: To achieve an optimum facial seal, the wearer should be clean-shaven. It is recommended the wearer be fit tested prior to use in accordance with AS/NZS 1715: 2009 and/or onsite requirements. Training on the safe use and limitations of the disposable face mask must be performed in accordance with AS/NZS 1715: 2009 section 2.4.
Box dimensions: 235mm x 100mm x 135mm
Box weight: 290g
Packaging dimensions: 240mm x 190mm x 120mm
Packaging weight: 190g (excluding contents)
For bulk orders over 500 masks please contact AusMed Health.